|Measurement Range||15 to 115 kPa (approximate)|
|Max. Pressure Exposure||400 kPa (exposure beyond this limit may cause permanent damage to sensor)|
|Sensitivity||45.9 mV per kPa; 0.459 mV per 0.01 kPa (approximate)|
|Calibration Factor||0.0218 kPa per mV (generic slope, reciprocal of sensitivity) and 11.4 kPa (generic intercept)|
|Measurement Uncertainty||± 1.5 kPa (with generic calibration coefficients)|
|Measurement Repeatability||Less than 0.1 %|
|Non-linearity||Less than 1 %|
|Warm-up Time||20 ms|
|Response Time||1 ms|
|Temperature Response||Less than 0.002 % per C for temperatures greater than 0 C; -0,015 % per C for temperatures less than 0 C|
|Operating Environment||-40 to 80 C; 0 to 100 % relative humidity (non-condensing)|
|Input Voltage Requirement||5 V DC|
|Output Voltage Range||0 to 5 V DC|
|Current Draw||7 mA DC|
|Dimensions||16 mm diameter|
Pressure is defined as force per unit area applied to a surface in a direction perpendicular to the surface. Barometric pressure or atmospheric pressure, is the force per unit area exerted on Earth's surface by the mass of air overlying the surface. High pressure indicates more atmospheric air mass over a given area, whereas, low pressure indicates less atmospheric air mass. Barometric pressure is strongly dependent on elevation, and decreases as elevation increases, due to less overlying air above the surface (shorter column of air) at higher elevations.
Barometric pressure is measured using a barometer. Aneroid (without liquid) barometers are often electronic and typically use capacitive elements to sensor pressure, with the major advantage of capacitive sensing mechanisms being minimal temperature dependence. Capacitive sensing circuits output a voltage that is related to pressure via sensor-specific calibrations. Apogee Instruments Barometric Pressure Sensor is an aneroid barometer consisting of a silicon capacitive sensing element and signal processing circuitry mounted in a compact epoxy plastic/stainless steel housing, and lead wires to connect the sensor to a measurement device.
• Barometric pressure measurement in weather networks, often for weather forecasting.
• Evapotranspiration calculations.
• Correct the output of sensors that are sensitive to press fluctuations (e.g., Apogee Instruments Oxygen Sensors).
If you would like to share your application of this product, please click here
Programs are in .CR1X format and can be downloaded for use with Campbell Scientific dataloggers. Right click and select "Save target as..." or an equivalent command in your browser. They can also be viewed using Wordpad or other text viewers.
Note: In 2020 the CR1000 Campbell Scientific datalogger was discontinued. Click here to access the discontinued .CR1 format sample datalogger programs >
Measurement of the Respiratory Quotient of Peat
Bowen Ratio Energy Balance Measurement of Carbon Dioxide (CO2) Fluxes of No-Till and Conventional Tillage Agriculture in Lesotho
A Short-Term Assessment of Carbon Dioxide Fluxes under Contrasting Agricultural and Soil Management Practices in Zimbabwe
Validation Experiments for Spent-Fuel Dry-Cask In-Basket Convertion